Glove Box Everyday Maintenance Guide
one) Fill in the use report in time immediately after running the glove box, like "user, water oxygen value, what to do"
two) When replacing gas cylinders in the vac glove box box, the stress is established to the default benefit ± 3mbar
3) The day by day doing the job pressure is set in accordance to the functioning comfort. When not in use for a long time, the force is established to 1-6mbar
4) Regularly test the dress in of the gloves, check the oil degree of the vacuum pump (amongst the upstream and downstream markings), and incorporate the pump oil properly
5) Clean up the water probe at the time just about every 2 to three months to make sure accurate detection. (Also need to have to refer to the actual scenario of the environment in various user packing containers
six) If you generally use organic solvents in the glove box, it is suggested that you regularly replace the natural solvent adsorbent substance, not afterwards than 1 yr at the latest
seven) Swap the filter element often, no later on than two years
eight) In accordance to the precise use of your glove box, decide to switch the vacuum pump oil cycle, this sort of as black and yellow pump oil, drinking water or far more organic and natural solvents in the pump oil. The oil replacement cycle should really not exceed two a long time at the most current
nine) If a big amount of organic solvent is inevitably exposed in the course of the experiment, the cycle demands to be stopped for the experiment. Right after the experiment, initial clean and then get started the cycle
ten) When the setting in the box is harmful or hazardous fuel, the exhaust port of the vacuum pump, the exhaust port of the regeneration gasoline and the exhaust port of the cleaning valve should be fatigued to the outdoor with an exhaust pipe (when exhaust is not allowed to be outdoors, the exhaust fuel procedure should be accomplished first).