Glove Box Day By Day Servicing Guide
1) Fill in the use history in time right after operating the vac glove box box, such as "user, drinking water oxygen worth, what to do"
2) When replacing gas cylinders in the glove box, the stress is set to the default benefit ± 3mbar
3) The each day doing work strain is established according to the working comfort and ease. When not in use for a lengthy time, the stress is established to one-6mbar
4) Routinely examine the use of the gloves, examine the oil degree of the vacuum pump (amongst the upstream and downstream markings), and include the pump oil correctly
5) Clean the water probe the moment every 2 to 3 months to make certain correct detection. (Also require to refer to the true predicament of the setting in distinctive user packing containers
6) If you often use organic and natural solvents in the glove box, it is advisable that you frequently replace the organic and natural solvent adsorbent content, not later than one 12 months at the newest
7) Exchange the filter factor frequently, no later on than two decades
8) In accordance to the actual use of your glove box, come to a decision to switch the vacuum pump oil cycle, these as black and yellow pump oil, water or far more organic solvents in the pump oil. The oil substitute cycle should not exceed 2 several years at the most recent
9) If a big amount of money of organic solvent is inevitably uncovered for the duration of the experiment, the cycle needs to be stopped for the experiment. Right after the experiment, very first clear and then start the cycle
10) When the surroundings in the box is toxic or dangerous gasoline, the exhaust port of the vacuum pump, the exhaust port of the regeneration gas and the exhaust port of the cleansing valve must be exhausted to the outside with an exhaust pipe (when exhaust is not authorized to be outdoors, the exhaust gas cure should be done to start with).